Gasman IECEx Certificate (Flammable model)
4. Certificate No.: Date of Issue: IECEx Certificate of Conformity IECEx BAS 05 . 0038 2014-11-04 Issue No . : 14 Page 4 of 4 DETAILS OF CERTIFICATE CHANGES (for issues 1 and above): Variation 14.1 To permit a minor change to the GASMAN Charger/Interface terminals not affecting the original assessment. Variation 14.2 I I , To permit the addition of an alternative component certified Gas Sensor for the GASMAN Flammable: SGX Sensortech (S . A.) Micro-Pellistor MP-7217 , MP-7218 andVQ548MP, covered by certificates IECEx ULD 14.0002U and l DEMKO 14 ATEX 1266U to standards IEC 60079-0 : 2011 Ed 6 & IEC 60079-11 :2011 Ed 6 and EN 60079-0 : 2012 Ed 6 & ; IEC 60079-11 :2012 Ed 6 respectively . ExTR: GB/BAS/ExTR14.0324/00 File Reference : 14/0615 I ' ~~~~~~~~~~~~~------------------~~~~~~~~~--------------------~ I
3. Certificate No . : Date of Issue: EQUIPMENT: IECEx Certificate of Conform ity IECEx BAS 05.0038 2014-11-04 Issue No. : 14 Page 3 of 4 Schedule Equipment and systems covered by this certificate are as follows : I The Gasman Flammable Gas Detector is designed to detect the p r esence of flammable gas, give a visual indication of ! ; the concentration, and give audible and visual alarms if a preset level is exceeded. It comprises a rechargeable battery, a flammable gas sensor head, electronic circuits on printed circuit boards, a liquid ' crystal display , LED's and a piezo sounder , all contained in a plastic enclosure providing a degree of protection of at least IP20 . i External contacts are provided for recharging, and for serial data communications . The flammable sensor used is either type VQ500 series by SGX Sensortech (IS) Ltd, to IECEx SIR 04 . 0014U, or 4P series , by City Technology to IECEx SIR 04 . 0013U , with a Code of Ex d IIC Gb and a maximum permitted ambient temperature of ' +55°C . These sensors are certified to EN60079-0 : 2009 and EN60079 - 1 :2007. The apparatus must only be recharged or connected to serial communications when in a non-hazardous area, using a ' Crowcon desktop charger or charger I interface part number C01940 or C01941. Variation 0.1 , The flammable gas sensor module is known as a flammable i-module . This variation permits the flammable i-module to be replaced by an oxygen i-module (for oxygen deficiency detection) , or a toxic i-module. The permitted i-modules are listed in the Use r Manual provided by the manufacturer. I The apparatus is not designed for use in oxygen enriched atmospheres . CONDITIONS OF CERTIFICATION: NO ----- -- -- ---_._--_ .. _._-----_. ! I ' I I I I I I I I
1. IECEx Certificate of Conform ity INTERNATIONAL ELECTROTECHNICAL COMMISSION IEC Certification Scheme for Explosive Atmospheres for rules and details of the IECEx Scheme visit www.iecex . com Certificate No.: Status: Date of Issue: Applicant: Electrical Apparatus : Optional accessory : Type of Protection : Marking : IECEx BAS 05 . 0038 issue No .: 14 Current 2014-11-04 Page 1 of 4 Crowcon Detection Instruments Ltd 172 Brook Drive Milton Park Abingdon Oxfordshire OX144SD United Kingdom Gasman Flammable Gas Detector Intrinsic safety, Flameproof Ex ia d IIC T4 (-20°C S Ta S +65°C) Approved for issue on behalf of the IECEx Certification Body : R S Sinclair Position : General Manager Signature: (for printed version) Date: 1. This certificate and schedule may only be reproduced in full. 2 . This certificate is not transferable and remains the property of the issuing body. Cert i ficate history: Issue No . 14 (2"014 · -11 : . 4) Issue No . 13 (2014-9 - 2) Issue No . 12 (2014-8-6) Issue No . 11 (2009-12- ! 17) Issue No . 1 0 (2008-8-6) Issue No . 9 (2008-6-16) Issue No . 8 (2008-4-4) Issue No . 7 (2007-7-5) Issue No . 6 (2006-12- 20) 3 . The Status and authenticity of this certificate may be verified by visiting the Official IECEx Website . Certificate i ssued by : SGS Baseefa Limited Rockhead Business Park Staden Lane Buxton Derbyshire SK17 9RZ United Kingdom
2. Certificate No.: Date of Issue : Manufacturer : Additional Manufacturing location (s): IECEx Certificate of Conform ity IECEx BAS 05.0038 2014-11-04 Crowcon Detection Instruments Ltd 172 Brook Drive Milton Park Abingdon Oxfordshire OX144SD United Kingdom Issue No.: 14 Page 2 of4 This certificate is issued as verification that a sample(s), representative of production, was assessed and tested and found to comply with the IEC Standard list below and that the manufacturer's quality system, relating to the Ex products covered by th i s certificate, was assessed and found to comply with the IECEx Quality system requirements . This certificate is granted subject to the conditions as set out in IECEx Scheme Rules, IECEx 02 and Operational Documents as amended. STANDARDS: The electrical apparatus and any acceptable variations to it specified in the schedule of this certificate and the identified documents, was found to comply with the following standards: IEC 60079-0 : 2011 Explosive atmospheres - Part 0 : General requirements Edition: 6.0 I EC 60079-1 : 2007-04 Explosive atmospheres - Part 1 : Equipment protection by flameproof enclosures "d" Edition : 6 lEe 60079-11 : 2011 Explosive atmospheres - Part 11: Equipment protection by intrinsic safety "i" Edition: 6 . 0 This Certificate does not indicate compliance with electrical safety and performance requirements other than those expressly included in the Standards listed above. TEST & ASSESSMENT REPORTS: A sample(s) of the equipment listed has successfully met the examination and test requirements as recorded in IECExATR: UKIBAS/04/0202 GB/BAS/ExTR14.0249/00 Gb/BAS/ExTR14.0324/00 File Reference: 04/0202 13/0719 14/0615
- 651 Total Views
- 562 Website Views
- 89 Embeded Views
- Social Shares
- 0 Likes
- 0 Dislikes
- 0 Comments
- 0 Facebook
- 0 Twitter
- 0 Google+
- 1 dsafety.com
D_S_Katalog_20155553 Views .
find det rigtige filter til dit åndedrætsværn4188 Views .
3M jupiter manual4075 Views .
D-S Produkt index3343 Views .
Guide til valg af åndedrætsværn3022 Views .
D-S Produktindex2998 Views .
D-S Product index2856 Views .
Skilte 20062801 Views .
speedglas_91002694 Views .
Åndedrætsværn 20152666 Views .
Airvisor 2 user manual - Multi languages2640 Views .
Procedure for iklædning af engangsbeklædning2522 Views .
Åndedrætsværn 20102352 Views .
Handsker størrelsesguide2180 Views .
EN 388 nye mærkninger forklaring2113 Views .
Handsker 20152112 Views .
Handsker 20122088 Views .
Ose.directory e-shot2048 Views .
Faldsikring 20101996 Views .
Handsker 20101976 Views .