Gasman IECEx Certificate (Oxygen and Toxic models)
4. Certificate No . : Date of Issue : IECEx Certificate of Conformity IECEx BAS 05.0039 2014-11-04 Issue No. : 14 Page 4 of4 DETAILS OF CERTIFICATE CHANGES (for issues 1 and above): -------- , Variation 14.1 I To permit a minor change to the GASMAN Charger/Interface terminals not affecting the original assessment. ExTR: GB/BAS/ExTR14.0324/00 File Reference: 14/0615 ! I I I : ' i I •
3. IECEx Certificate of Conformity Certificate No.: IECEx BAS 05 . 0039 Date of Issue: 2014-11-04 Issue No .: 14 Page 3 of 4 Schedule EQUIPMENT: Equipment and systems covered by this certificate are as follows: , The Gasman Oxygen or Toxic Gas Detector is designed to detect the presence of oxygen (deficiency) or toxic gas, give a visual indication of the concentration, and give audible and visual alarms if a preset level is exceeded. It comprises a battery, a gas sensor head, electronic circuits on printed circuit boards, a liquid crystal display, LED's and a piezo sounder, all contained in a plastic enclosure providing a degree of protection of at least IP20 . Two types are available, one rechargeable, and one non-rechargeable . External contacts are provided for recharging , and for serial data communications. , The sensors are known Oxygen or Toxic i-modules, and the permitted i-modules are listed in the User Manual provided by the manufacturer. ' The apparatus must only be recharged or connected to serial communications when in a non-hazardous area , using a Crowcon desktop charger or charger / interface part number C01940 or C01941 . In the case of the non-rechargeable apparatus , the charging circuit will have no effect. The cell in the non-rechargeable apparatus must only be changed when in a non-hazardous area, and replaced only with , cell type CR24 77. The apparatus is not designed for use in oxygen enriched atmospheres . CONDITIONS OF CERTIFICATION: NO i I I I
1. IECEx Certificate of Conform ity INTERNATIONAL ELECTROTECHNICAL COMMISSION IEC Certification Scheme for Explosive Atmospheres for rules and details of the IECEx Scheme visit www . iecex . com Certificate No . : Status : Date of Issue : Applicant: Electrical Apparatus: Optional accessory : Type of Protection: Marking: IECEx BAS 05 . 0039 issue No . :14 Current 2014-11-04 Page 1 of 4 Crowcon Detection Instruments Ltd 172 Brook Dr i ve Milton Park Abingdon Oxfordshire OX144SD United Kingdom Gasman Oxygen or Toxic Gas Detector Intrinsic Safety Ex ia IIC T4 Ga (_20DC::S Ta::S +65 D C) Approved for issue on behalf of the IECEx Certification Body: R S Sinclair Position : General Manager Signature: (for printed version) Date : 1 . This certificate and schedule may only be reproduced in full. 2. This certificate is not transferable and remains the property of the issuing body. Certificate history : Issue No . 14 (20 1 4-11- 4) Issue No. 13 (2014 - 9 - 2) Issue No . 12 (2014 - 8-6) . Issue No . 11 (2009-12- I 17) Issue No . 1 0 (2008-8-6) Issue No . 9 (2008 - 6 - 16) , Issue No . 8 (2008-4-4) Issue No . 7 (2007 - 7 - 5) Issue No . 6 (2006-12 - I 20) 3. The Status and authenticity of this certificate may be verified by visiting the OfficiallECEx Website . Certificate issued by : SGS Baseefa Limited Rockhead Business Park Staden Lane Buxton Derbyshire SK179RZ United Kin gdom
2. Certificate No.: Date of Issue: Manufacturer : Additional Manufacturing location (s) : IECEx Certificate of Conformity IECEx BAS 05.0039 2014-11-04 Crowcon Detection Instruments Ltd 172 Brook Drive Milton Park Abingdon Oxfordshire OX144SD United Kingdom Issue No. : 14 Page 2 of 4 This certificate is issued as verification that a sample(s), representative of production, was assessed and tested and found to comply with the IEC Standard list below and that the manufacturer's quality system, relating to the Ex products covered by this certificate , was assessed and found to comply with the IECEx Quality system requirements. This certificate is granted subject to the conditions as set out in IECEx Scheme Rules , IECEx 02 and Operational Documents as amended. STANDARDS: The electrical apparatus and any acceptable variations to it specified in the schedule of this certificate and the identified documents, was found to comply with the following standards: IEC 60079-0 : 2011 Edition : 6 . 0 IEC 60079-11 : 2011 Edition : 6.0 Exp l osive atmospheres - Part 0 : General requirements Explosive atmospheres - Part 11: Equipment protection by intrinsic safety "i" This Certificate does not indicate compliance with electrical safety and performance requirements other than those expressly included in the Standards listed above. TEST & ASSESSMENT REPORTS: A sample(s) of the equipment listed has successfully met the examination and test requirements as recorded in IECExATR: UKIBAS/04/0202 GB/BAS/ExTR14/0249/00 GB/BAS/ExTR14/0324/00 File Reference: 04/0202 13/0719 14/0615
- 750 Total Views
- 660 Website Views
- 90 Embeded Views
- Social Shares
- 0 Likes
- 0 Dislikes
- 0 Comments
- 0 Facebook
- 0 Twitter
- 0 Google+
D_S_Katalog_20155580 Views .
find det rigtige filter til dit åndedrætsværn4205 Views .
3M jupiter manual4085 Views .
D-S Produkt index3350 Views .
Guide til valg af åndedrætsværn3043 Views .
D-S Produktindex3014 Views .
D-S Product index2864 Views .
Skilte 20062815 Views .
speedglas_91002711 Views .
Åndedrætsværn 20152679 Views .
Airvisor 2 user manual - Multi languages2661 Views .
Procedure for iklædning af engangsbeklædning2538 Views .
Åndedrætsværn 20102360 Views .
Handsker størrelsesguide2197 Views .
Handsker 20152120 Views .
EN 388 nye mærkninger forklaring2120 Views .
Handsker 20122096 Views .
Ose.directory e-shot2053 Views .
Faldsikring 20102005 Views .
Handsker 20101980 Views .